FREND™ PSA Plus
Prostate Specific Antigen
1. Intended Use
The FREND™ PSA Plus is designed for in vitro DIAGNOSTIC USE ONLY for the quantitative measurement of
total Prostate Specific Antigen (PSA) in human serum, heparinized plasma, and EDTA plasma using the FREND™
System. This device is indicated for the serial measurement of total PSA in serum, heparinized plasma and EDTA
plasma to be used as an aid in the management of patients with prostate cancer.
2. Summary and Explanation of Test
Prostate-specific antigen (PSA) is a single-chain glycoprotein with molecular weight of 34 kilodaltons.1,2,3 As a
serine protease with chymotrypsin-like activity, PSA belongs to the kallikrein family. In blood, PSA exists as a free
or complex form with protease inhibitors such as α-1-antichymotrypsin (ACT). Total PSA represents the sum of
both free and complex forms.4 PSA is uniquely associated with prostate tissues from normal, inflamed or cancerous
stages. Elevated PSA in serum or plasma is found in patients with prostate cancer, benign prostatic hypertrophy, or
inflammatory tissues. Studies on a variety of PSA methods have shown that PSA can be useful as an indicator for
the diagnosis and management of prostate cancer.5
PSA has been found in normal, benign hyperplastic, and malignant prostatic tissue, in metastatic prostatic carcinoma,
and also in prostatic fluid as well as in seminal fluid.6 PSA is not found in any other tissue in men, and it is not
produced by cancers originating in the lung, colon, rectum, stomach, pancreas or thyroid.7 Though increased
concentrations of PSA are found in the serum of patients with benign prostate hyperplasia (BPH), prostatitis and
prostate infections and inflammation, they are also found in patients with cancer of the prostate.8,9 PSA measurement
is an essential tool in assessing the status of disease in patients with prostate cancer when serial samples are
measured over time.10 The clinical value realized by monitoring tPSA concentrations in patients with prostate cancer
regardless of the treatment regimen is well known.11 Since the mid-1980’s, there has been a growing body of
literature concerning the utility of Prostate Specific Antigen (PSA) for both the monitoring and detection of prostate
3. Principle of the Assay
The FREND™ PSA Plus is a rapid quantitative “sandwich” immunoassay using fluorescent nanoparticles which
measures the concentration of total PSA. Thirty μL of patient serum or plasma (heparin or EDTA only) is manually
presented to the inlet on the individual single-unit test cartridge where it is mixed with fluorescent nano-particles
conjugated with PSA antibodies. PSA molecules in the specimen bind to conjugated antibodies to form immune
complexes which then move by capillary action through the reagent cartridge channel to the detection area. When
the specimen reaches the test zone, it hydrates dried solid-phase anti-PSA antibodies. PSA-fluorescent particleimmune complexes in the specimen are grabbed by the capture antibodies to form sandwich immune-complexes.
The residual PSA-unbound fluorescent nano-particles conjugated with PSA-antibodies pass through the test zone
and bind to PSA antigens in the reference zone. As the samples moves forward to the waste reservoir, non-specific
binding components are washed away. The intensity of fluorescence measured by a light source (laser) is
proportional to the amount of total PSA in the original sample. The result is calculated using information stored on
the lot specific FREND™ PSA Plus Code chip and then is displayed on the FREND™ System screen. A hard copy
printout can be obtained if desired. A ratio calculated between the Reference zone and the Test zone corrects for
Total PSA concentration in a sample analyzed with the FREND™ PSA Plus on the FREND™ System correlates
directly with the fluorescence intensity- the higher the tPSA concentration, the greater the fluorescence. The
FREND™ PSA Plus has a measuring range determined as 0.1ng/mL to 25.0ng/mL.
The FREND™ PSA Plus uses single-use transparent plastic cartridges in which all required reagents are stored
within the cartridge itself. All that is added by the user is a 30μL test sample. The cartridge is inserted into the
FREND™ System in a prescribed fashion indicated with a black arrow on the cartridge. The reaction is read multiple
times as the sample moves via capillary action through the cartridge. This type of assay system is sometimes referred
to as one which incorporates laminar flow.
4. Material Provided (FREND™ PSA Plus)
FREND™ PSA Plus cartridges
Disposable pipette tips (micro-pipettor provided)
FREND™ PSA Plus Code chip
FREND™ PSA Plus Package Insert
One Cartridge contains:
5. Materials Required But Not Provided
The following materials are not provided with the cartridge but are required to perform Prostate Specific Antigen
analysis using the FREND™ PSA Plus on the FREND™ System. They are available separately from NanoEnTek.
6. Warnings and Precautions
The FREND™ PSA Plus cartridges are intended for in vitro diagnostic use only.
PSA Plus cartridges are only to be used on the FREND™ System.
Allow cartridges to come to room temperature for 15~30 minutes prior to use.
Avoid cross-contamination between samples by using a new pipette tip for each new specimen.
Avoid high humidity, direct sunlight or heat in the area used for cartridge storage.
Inaccurate results are possible if the sample used is contaminated in any way.
Using specimens containing clotted fibrin could result in erroneous results.
Over or under loading the cartridge with sample may result in inaccurate results.
Cartridges should not be frozen.
Human specimens are not used in the preparation of this product, however, since human specimens will be used for
samples and other quality control products in the lab may be derived from human materials, please use standard
laboratory safety procedures when handling all specimens and controls.
Do not use the cartridges beyond the expiration date on the pouch.
Do not use the cartridge if the pouch is damaged or the seal is broken.
Perform testing as specified in the Package Insert and User Manual.
PSA Plus cartridges are disposable, single use devices. Do not reuse them under any circumstances.
Keep the cartridge sealed in the pouch until just ready for use.
Use the cartridge immediately after opening its pouch.
Wear disposable gloves when handling the cartridges and the samples.
Wash hands thoroughly and often handling reagent cartridges or samples.
PSA Plus has been designed so that the high dose “hook effect” is not a problem for the vast majority of samples.
Samples with PSA concentration between 25 and 1,200 ng/mL will read > 25 ng/mL. The “hook effect”
phenomenon may occur only at PSA concentration > 1,200 ng/mL.
7. Storage and Stability
All unopened materials are stable until the expiration date on the label when stored at the specified temperature.
Cartridge stability has been demonstrated for twelve months from the date of manufacture.
The expiration date is clearly indicated on the product box and the cartridges.
FREND™ PSA Plus cartridges
Refrigerator temperature (2~8 ℃)
Room temperature (18~25 ℃)
8. Specimen Collection and Handling
Serum or plasma (heparinized or EDTA only) is required for the assay. Citrated plasma SHOULD NOT BE USED.
No special patient preparation is necessary. To use serum, a blood sample is collected aseptically without additives
by venous puncture. After allowing the sample to clot for 30 minutes at room temperature, the collection tube should
be centrifuged for 10 minutes at 3,000 rpm.
For heparinized or EDTA plasma, a venous blood sample is collected aseptically with the designated additive. The
plasma should be separated from the packed cells as soon as possible.
Prostatic manipulation has been shown to affect the PSA results so samples should be drawn before any prostatic
procedures such as DRE, prostatic massage and TRUS are performed.
Samples may be stored at 2~8 ℃ for up to 6 hours prior to analysis. If the analysis is scheduled to be done at some
later time, the sample should be stored frozen at -20 ℃ or below for future use. Sample stability study was
performed at -20 ℃ only for three weeks by manufacturer but there are many literature available showing that
total PSA kept frozen is stable for 3 months or longer.16, 17, 18
Repeated freeze-thaw cycles should be avoided. Turbid serum samples or samples containing particulate matter
such as fibrin clots or strands should be centrifuged before being tested. Prior to assay, slowly bring frozen samples
to room temperature (18~25 ℃) and mix gently but thoroughly before testing.
1) Reagent Preparation
There is no reagent preparation required to measure tPSA using the FREND™ PSA Plus cartridge on the
FREND™ System. However, the cartridges needed for a particular run should be removed from the
refrigerator and allowed to reach room temperature for 15~30 minutes before they are used.
The calibrators used during the cartridge manufacture process to create the information placed
electronically on the FREND™ PSA Plus Code chip are prepared gravimetrically and are compared to
international reference standards (WHO International Prostate Specific Antigen (90:10) NIBSC code:
96/670). However, for the end user, there is no need for calibration as is generally performed on other
automated laboratory equipment. All calibration statistics and information have been electronically stored
on the FREND™ PSA Plus Code chip included in each box of FREND™ PSA Plus cartridge. The
FREND™ PSA Plus Code chip is specific for that manufactured lot of FREND™ PSA Plus.
T The appropriateness of the calibration information should always be checked by running sufficient
external quality control materials as samples to verify that the results obtained for tPSA on the FREND™
System using the FREND™ PSA Plus cartridges of a particular lot met the laboratory criterion for
ㆍPSA Plus Code Chip Installation
Please refer to the FREND™ System User Manual for more detailed instructions relative to the Code chip
installation. Abbreviated instructions follow here:
Insert the FREND™ System electrical cord into an appropriate outlet.
Insert the Code chip into the code chip slot at the rear of the FREND™ System following the arrows.
Press the ‘Setup’ button on the ‘Main’ screen.
Press the ‘Code chip’ button on the ‘Setup’ screen.
The information embedded on the FREND™ PSA Plus Code chip is automatically saved on the
When the Code chip installation is completed, press the ‘OK’ button to go to the ‘Setup’ screen.
Press the ‘Item’ button on the ‘Setup’ screen.
Check the FRND™ PSA Plus cartridge lot number and the installation date of the Code chip.
Press the ‘Home’ button to go to the ‘Main’ screen to begin running external quality control and patient
3) Quality Control
ㆍCommercially Available Controls
C Commercially available controls from a variety of manufacturers are available that contain tPSA as a
measured analyte. It is recommended that these external controls be run at least once per day when testing
is scheduled for FREND™ PSA Plus on the FREND™ System.
A minimum of at least two (2) levels of controls, normal and abnormal, should be used.
Individual laboratory policy will dictate exactly which control materials and lot numbers should be run, the
frequency with which controls are to be tested, criteria for acceptance of the results and required corrective
action to be taken if results do not meet laboratory criteria.
Do not assay patient samples on the FREND™ System using the FREND™ PSA Plus if quality control
results do not give expected values. Refer to your laboratory policies on how to determine acceptability of
external control material results.
ㆍQuality Control Procedure
External quality control materials to be assayed using the FREND™ PSA Plus is defined by individual
laboratory policy. The assay procedure used to obtain results on the selected external quality control
material is identical to that used when testing patient samples and follows below.
If lyophilized quality control material is to be used, please reconstitute according to the manufacturer’s
instructions and allow the pellet to dissolve as stipulated until the material has gone into solution. Be sure
to mix gently but thoroughly before testing.
Refrigerated and/or frozen control material should be treated as described below under preparation for
specimen processing. All materials should be at room temperature before use.
4) Specimen Processing
Remove from the refrigerator sufficient cartridge of FREND™ PSA Plus to test the number of patient
samples and required external quality control materials. Allow the cartridges to come to room temperature
for 15~30 minutes prior to the start of the testing sequence.
If using refrigerated patient samples, remove those from the refrigerator and allow to them to come to room
temperature prior to testing. If frozen samples will be utilized, be sure these are removed from the freezer,
thawed naturally and then mixed gently but thoroughly prior to testing
There are no other reagents or sample preparations necessary.
(1) Prepare the FREND™ PSA Plus and specimen.
(2) Record the Sample ID on the cartridge in the designated area.
(3) Drop the sample (30μL) into the sample inlet on the cartridge using the FREND™ System pipettor
with a fresh pipette tip.
(4) Press the ‘Test’ button on the ‘Main’ screen of the FREND™ System.
(5) The screen of FREND System moves to the Patient ID screen automatically.
(6) Type the Patient ID and press the ‘Enter’ button to begin the test.
(7) Insert the cartridge into the cartridge slot using the cartridge arrows as a guide.
Caution: Please check the direction of the cartridge before insertion and assure the insertion is complete.
(8) When the reaction in the cartridges is complete in 6 minutes, the FREND™ System will automatically
begin the reading process.
(9) When the measurements are completed, the cartridge will automatically be expelled and the results
Caution: Do not remove power from the FREND™ System while a cartridge is in the reading chamber.
This may cause a system error.
If the FREND™ System is connected to the optional printer, press the ‘Print’ button and the
results will be output on the printer paper.
For more detailed instructions, please refer to the FREND™ System User Manual.
10. Procedural Notes
If a specimen Prostate Specific Antigen concentration is found to be greater than the linearity limit of the assay of
25.0 ng/mL and a definitive result is required, the specimen should be diluted with female sera that has been
previously measured on the FREND™ PSA Plus and found to contain < 0.1 ng/mL tPSA and then re-assayed
according to the Assay Procedure. The recommended dilution for samples with an initial result of >25.0 ng.mL is
1:10 or 1:50. It is desirable to dilute the sample so that the diluted sample reads between 2 and 20 ng/mL. Dilutions
must be made manually and the final result on the diluted sample calculated manually by multiplying the result
obtained on the diluted sample by the dilution factor.
11. Calculation of Results
The FREND™ System performs all sample and cartridge handling operations automatically within the cartridge
once the sample has been manually added to the sample well in the cartridge and the cartridge placed into the
FREND™ System. The rate of fluorescence produced by the reaction is read at various intervals during the analysis
process, blank readings are subtracted after which the net rate is automatically converted to total Prostate Specific
Antigen concentration in ng/mL based upon information stored on the PSA Code chip. This result is then output on
the screen and to the optional printer. It is also stored in memory on the FREND™ System.
Screen Displays for Various Concentration Scenarios
12. Evaluation of Results
In order to monitor and evaluate the precision of the analytical performance, it is recommended that commercially
available control samples be assayed daily.
The minimum recommendations for the frequency of running internal control material are:
ㆍ When beginning a new lot of cartridge, three levels of controls are run in order to validate the calibration.
ㆍ The three levels of controls are also repeated when certain service procedures are performed such as optical
adjustment or change.
If any external quality control sample values are out of the acceptable range, it will be necessary to investigate the
problem before reporting patient results to assure there is not an instrument or software malfunction.
Each laboratory operates under a different set of regulations. Every laboratory must follow the standardized
procedures acceptable to the regulatory agencies to whom the laboratory is responsible.
13. Limitations of the Procedure
When used for diagnostic purposes, the results obtained from this assay should be used in conjunction with other
data(e.g. symptoms, results of other tests, clinical impressions, medical history, therapy, etc.).
The FREND System, paired with a FREND PSA Plus cartridge, is programmed to report 25.0 ng/mL as the highest
concentration of PSA measurable without dilution. The lowest measurable concentration is 0.1 ng/mL – the assay
Heterophilic antibodies in a sample have the potential to cause interference in immunoassay system12, 13.
Infrequently, PSA level may appear elevated due to heterophilic antibodies present in the patient’s serum or plasma
or to nonspecific protein binding. If the PSA level is inconsistent with clinical evidence, additional PSA testing is
suggested to confirm the results.
Although hemolysis has an insignificant effect on the assay, hemolyzed samples may indicate mistreatment of a
specimen prior to assay and results should be interpreted with caution.
Lipemia has an insignificant effect on the assay except in the case of gross lipemia where interference with the
lateral flow of the sample in the cartridge may occur.
Specimens from patients who have received preparations of mouse monoclonal antibodies for diagnosis or therapy
may contain human anti-mouse antibodies (HAMA). Such specimens may show falsely elevated or decreased PSA
Certain medications may interfere with assay performance. All results should be interpreted with respect to the
clinical picture of the patient14.
The concentration of tPSA in a given sample determined with assays from different manufacturers can vary due to
differences in assay methods, calibration, and reagent specificity15.
Please refer to the Specimen Collection and Handling, Warnings and Precautions, Storage and Stability, and
Procedural Notes sections in this insert sheet.
Clinical results must be interpreted with regard to medications administered to the patient14.
The ability of the assay to detect both free and complexed forms of total PSA (free PSA complexed with alpha-1antichymotrypsin) on an equal molar basis (equimolarity) has not been established.)
14. Expected Values
As with every clinical diagnostic test, a reference interval corresponding to the characteristics of the population
being tested should be determined by each laboratory. The FREND™ PSA Plus on the FREND™ System is to be
used on serial blood samples to manage patients with prostate cancer
Testing of ambulatory male subjects fifty years old and older who reported themselves as healthy without any known
illnesses, diseases or conditions was performed using both the FREND™ PSA Plus on the FREND™ System and
another commercially available PSA method. The currently accepted reference interval for tPSA of up to 4.0 ng/mL
was validated for both systems.
As is true for all PSA methods, no tPSA results can be interpreted as being definitive for the presence or absence of
prostate cancer. Patients with levels of PSA within the reference interval found in apparently healthy subjects may
have prostate cancer; patients with levels exceeding those in the reference interval may be prostate cancer free.
Results from the FREND™ PSA Plus on the FREND™ System should be interpreted in the light of other clinical
findings and diagnostic procedures such as DRE, various imaging studies, etc. since certain treatments can cause
PSA values to decrease by virtue of the treatment while the cancer is still progressing.
15. Reference Ranges
The interval given here was determined in serum samples from 196 apparently healthy male subjects from the age
of 50~71 years.
Number of samples (n)
In this study, greater than 99% of the healthy subjects had serum PSA concentrations less than or equal to 4.0 ng/mL
by FREND™ PSA Plus on both the FREND™ System and another commercially available tPSA fluorescent assay.
Results of this study are shown below.
Number of samples (n)
Range of values
Other FDA cleared PSA
PSA Plus on FREND™ System
Expected Values for Management of Patients with Prostate Cancer
Distribution of Serum FREND™ PSA Plus concentrations
Healthy, Benign and Various Malignant Disease States
Men ≥ 50 yrs.
Gleason Score 5~6
Gleason Score 7
Gleason Score 8~9
GB, Gastric, Pancreatic
*Treated and untreated subjects
**Serial samples are not included in this cohort.
16. Performance Characteristics
Performance characteristics were evaluated for the FREND™ PSA Plus as follows:
Known samples of PSA were added to a female serum specimen (0.01 ng/mL). The concentration of PSA
was determined before and after the addition of the exogenous PSA and the percent recovery was calculated.
Spiked Recovery FREND™ PSA Plus
1b) Dilution Linearity
Specimens from a high concentration pool (34 ng/mL tPSA) were diluted with a low concentration pool
following instructions in the CLSI EP6-A document. Correlation with the expected linearity showed
Dilution Linearity - FREND™ PSA Plus
Dilution Linearity Data - FREND™ PSA Plus
1c) Comparative Analysis
A group of well-characterized serum samples collected with IRB oversight from subjects with pathologyverified prostate cancer and stored at -70 ℃ under monitored conditions were analyzed for total PSA.
Results from the FREND™ PSA Plus on the FERND™ System (y) were compared to those obtained using
the Tosoh ST AIA-PACK PA assay (x). A total of 160 unique samples were analyzed in the study, however,
only samples with tPSA results within the linearity of the FREND™ PSA Plus (up to 25.0 ng/mL; (n=143))
were used in the comparative analysis.
Slope: 0.9192 (95% CI: 0.8369; 1.0014)
Intercept: -0.01179 (95% CI: -0.2763; 0.2527)
Correlation Coefficient (R) = 0.9671 (95% CI: 0.9545~0.9763)
Number of samples = 143
Range of FREND™ PSA Plus values: 0.04~29.99 ng/mL
Range of ST AIA PA values: 0.00~25.86 ng/mL
2a) Precision Testing Single Lot and Single Site
Precision was determined as described in the CLSI protocol EP5-A. Three clinical samples across the
measuring range were assayed in replicates of two at two separate times per day for twenty days using a
single lot of FREND™ PSA Plus cartridge. The findings follow showing repeatability, between-run,
between-day, and within-laboratory precision data.
2b) Precision Testing Multiple Lots and Multiple Sites
Three different lots of FREND™ PSA Plus were evaluated at three geographically diverse sites. Four
replicates each of Material A, Material B, and Material C and two replicates of QC 1, QC 2 and QC 3 were
evaluated in each of two runs performed for five days at each site. A total of 40 results on each material
were generated at each of the three sites yielding a grand total of 120 replicates of each material. The data
was analyzed using a CLSI format from EP5-A2 for an ANOVA analysis. Instrument-to-Instrument is the
same as Site-to-Site in this scenario. As can be seen from the table below, the largest source of variation is
the cartridge which would be the expected result. The FREND™ PSA Plus cartridge is a single use cartridge
that contains all the reagents within the cartridge necessary to support the reactions.
CV% by Material
The following substances were evaluated for potential cross-reactivity with the FREND™ PSA Plus at the
concentrations indicated below. Testing was done according to the instructions recommended by CLSI
protocol EP7-A. No significant cross-reactivity was found.
Specificity of FREND™ PSA Plus
(Prostatic Acid Phosphatase)
4) Analytical Sensitivity
The Limit of Detection (LoD) for the FREND™ PSA Plus was determined using the CLSI EP17-A protocol.
The analytical sensitivity of the FREND™ PSA Plus was determined to be 0.1 ng/mL.
Interference is defined, for purposes of this study, to be recovery outside of 15% of the known specimen mean
concentration. In other words, recovery from 85% to 115% of the expected is considered acceptable performance.
1) Endogenous Interference
These interference studies on endogenous substances were performed using the FREND™ PSA Plus on
the FREND™ System according to the recommendations in the CLSI protocol EP7-A:
Added hemoglobin (up to 500 mg/dL) does not interfere with the assay. Average recovery when added to
serum containing tPSA at 10 and 4.0 ng/mL was 97.25%.
Added conjugated bilirubin (up to 20 mg/dL) does not interfere with the assay. Average recovery when
added to serum contaiing tPSA at 1.0 and 4.0 ng/mL was 98.2%.
Added gamma globulin (Total Protein) up to 5.0 g/dL does not interfere with the assay. Average recovery
when added to serum containing tPSA at 1.0 and 4.0 ng/mL was 106.3%.
Added triglyceride up to 3 g/dL does not interfere with this assay. Average recovery when added to serum
containing tPSA at 1.0 and 4.0 ng/mL was 101.5%.
2) Pharmaceutical Interference
The following chart shows the interference studies performed using the FREND™ PSA Plus on the
FREND™ system for various drugs that might be found in the serum/plasma of men diagnosed with
prostate cancer. The concentrations of the drugs that were added to the test samples and controls at base
concentration of tPSA of 1.0 ng/mL and 4.0 ng/mL are shown as are the % recoveries on the far right side
of the chart. It is unlikely since the FREND™ PSA Plus method uses monoclonal antibodies that any
substances without a tertiary structure similar to PSA would interfere. The testing showed that indeed there
was no significant interference from the tested drugs that would affect the interpretation of a tPSA result
as assayed on the FREND™ PSA Plus
Interference Study Results for FREND™ PSA Plus on the FREND™ System
Average Recovery %
18. Serial Measurements and concordance with medical Status
Since the FREND™ PSA Plus indication for use is that the assay results will be used as a tool in managing the care
for patients with prostate cancer, it is imperative that the changes in the marker are compared to clinical status
changes to determine the efficacy of the test. Therefore, as an important part of the clinical studies performed to
characterize the FREND™ PSA Plus, serial samples collected longitudinally from patients previously diagnosed
with prostate cancer and treated in a variety of ways over the clinical course of their disease (including prostatectomy,
radioactive seeds, external beam radiation, chemotherapy, hormone therapy alone or in combination) were assayed
for tPSA with the FREND™ PSA Plus on the FREND™ System. The same samples were also measured for tPSA
by another FDA cleared method.
For each point to point in as sample serial set, the change in the tPSA concentration was compared to the change in
the clinical status of the patients as measured by other laboratory tests, patient interviews, physical examinations,
and imaging studies of a variety of types and recorded on a Clinical Report Form.
These changes in the tPSA marker concentration were defined as significant or not by multiplying the overall CV
of the assay at the midrange (as determined by the test imprecision study) by a factor of 2.5 to define a percentage
change different from what would be expected because of assay imprecision. For the FREND™ PSA Plus assay
with an overall mid-range CV of 8.5%, significance was set at a change in excess of 20%. Any increase in value
from one time period to the next that did not exceed 20% was logged as ≤ 20% change. For the other FDA cleared
method, significance was set at a change > 8.5%. This was calculated using that method’s published overall midrange CV of 3.4% x 2.5.
Physician’s impressions regarding subject’s disease status at each blood draw were recorded on CRFs for all serial
PSA samples. The status information at one visit was compared with the status at the next and a change in status
was determined for the visit pair. The change in clinical status from the physician’s impression was then compared
to significant changes in the PSA concentration of each sample as measured by both the test device (FREND™ PSA
Plus) and the predicate device (FDA Cleared Assay). The results of these comparisons are shown below where
clinical status was divided into two groups: those visit pairs showing progression and those showing no disease
progression per the visit to visit status.
1) Samples with no Progression (Negative Concordance)
FREND™ PSA Plus
Other FDA Cleared PSA Assay
NC(FREND™ PSA Plus)=
There is no difference in the Negative Concordance (NC) between the two assays.
2) Samples with Progression (Positive Concordance)
FREND™ PSA Plus
Other FDA Cleared PSA Assay
NC(FREND™ PSA Plus)=
A McNemar test of the paired data in the above table yields a p value of 0.218. The associated 95%
confidence interval for the true difference is -0.0157 to 0.0551. Both estimators agree: there are no
statistically significant differences in the Positive Concordances (PC) between the two assays.
Below is chart comparing the concordances of the FREND™ PSA Plus assay and the other FDA cleared
PSA assay. Based on the 95% confidence intervals there appears to be no differences between the
concordances for the FREND™ PSA Plus assay and the other FDA cleared PSA assay.
69.8% to 85.3%
55.1% to 71.5%
64.1% to 76.9%
*Confidence intervals are based on 10,000 resamples of the patient data
73.85% to 88.7%
55.1% to 71.5%
66.1% to 77.5%
de Koning HJ, Liem MK, Baan CA, Boer R, Schröder FH, Alexander FE, Prostate cancer mortality reduction
by screening: power and time frame with complete enrollment in the European Randomized Screening for
Prostate Cancer (ERSPC) trial. International Journal Cancer, 2002; 98:268-73.
Wan-Ming Zhang, Patrik Finne, Jari Leinonen et al., Characterization and immunological determination of
the complex between prostate-specific antigen and a2-macroglobulin, Clinical Chemistry, 1998, 44:12,
David L. Woodrum et al., Analytical performance of the Tandem-R free PSA immunoassay measuring free
prostate-specific antigen. Clinical Chemistry, 1997, 43:7, 1203–1208.
Yun Sik Kwak, M.D., Appropriate use of Prostate-Specific Antigen in Diagnosing Carcinoma of the Prostate.
Journal Korean Geriatrics Society, 2009, Vol.13, Mar, pp. 12-17.
Chen Z, Prestigiacomo A, et al., Purification and characterization of prostate-specific antigen(PSA)
complexed to alpha 1-antichymotrypsin: potential reference material for international standardization of PSA
immunoassays, Clinical Chemistry, 1995, 41(9):1273-82.
Wang, M.C., et al, 1981, Prostatic Antigen: A New Potential Marker for Prostatic Cancer. Prostate 2:89.
Frankel, A.G., et al, 1982, Monoclonal Antibodies to a Human Prostate Antigen. Cancer Res. 42:3714
Papsidero, L.D., et al, 1980, A Prostate Antigen in Sera of Prostatic Cancer Patients. Cancer Res. 40:2428.
Kuriyama, M., et al, 1980, Quantitation of Prostate-Specific Antigen in Serum by a Sensitive Enzyme
Immunoassay. Cancer Res. 40:4658.
Killian, C.S., 1985, Prognostic Importance of Prostate-Specific Antigen for Monitoring Patients with Stages
B2 to D1 Prostate Cancer. Cancer Res. 45:886.
Kuriyama, M., et al, 1982, Multiple Marker Evaluation in Human Prostate Cancer with the Use of TissueSpecific Antigens. J. Nat. Canc. 68:99.
Nahm MH and Goffman JW. Heteroantibody: Phantom of the Immunoassay. Clinical Chemistry, 1990, 36:829.
Boscato LM., Stuart MC., Heterophilic Antibodies: A Problem for All Immunoassay Compared. Clinical
Chemistry, 1988, 33:1916-20.
Young, D., 1990, Effects of Drugs on Clinical Laboratory Tests. 3rd Edition, Washington, DC, American
Association for Clinical Chemistry Press.
Chan DW, Bruzek DJ, Oesterling JE, et al., Human Anti-Murine Immunoassays, Clinical Chemistry, 1997,
Chemistry Analyte Stability Chart, Barnes Jewish Hospital, Apr 1999, Revised on Jul 2011.
Melissa Tanner, Neil Kent, Brian Smith, Stephen Fletcher and Michelle Lewer, Stability of common
biochemical analytes in serum gel tubes subjected to various storage temperatures and times pre-centrifugation,
Annals of Clinical Biochemistry, 2008, 45, 375-379.
18. G. Rajani Kumari and T. Malati, Stability of total and free prostate specific antigen in serum samples at different
storage conditions, Indian Journal of Clinical Biochemistry, 2004, 19 (2), 10-13.
20. Glossary of Symbols
Do not reuse
Use by YYYY-MM-DD
Warning or Caution
Authorized representative in the Europe Community
In vitro diagnostic medical device
Contains sufficient for <n> tests
Email: [email protected]
NanoEnTek Inc. (H.Q.)
12F, 5, Digital-ro 26-gil, Guro-gu, Seoul, 152-740, Korea
Tel. +82-2-6220-7940 / Fax. +82-2-6220-7721
851-14, Seohar-ro, Paltan-myeon, Hwaseong-si, Gyeonggi-do, 445-917, Korea
NanoEnTek USA Inc.
5627 Stoneridge Drive Suite 304, Pleasanton, CA 94588, USA
Tel. +1-925-225-0108, +1-888-988-0108(toll free) / Fax. +1-925-225-0109
SK Medical (Beijing) Co., Ltd.
26F, SK Tower, No. 6 Jia, Jianguomenwai Avenue, Chaoyang district, Beijing, 100022, P.R. China
Tel. +86-10-5920-7844 / Fax. +86-10-5920-5697
MT Promedt Consulting GmbH
Altenhofstrasse 80, 66386 St. Ingbert, Germany